ambient to +45.0°C
Nade Lab Testing Equipment Automatic Tablet/capsule Dissolution Tester RC-6
RC series of instruments are used for detecting velocity and extent of pharmic dissolution from tablet, capsule etc in prescriptive menstruum.
National standard( Chinese codex ) /2010
Corporate standard(Dissolution tester )Q/12XQ0187-2010
It has six vessels and six poles arranged on one line.
The head part of the instrument can be artificially turned over smoothly and flexibly.
The basket units and paddle units are made of imported stainless steel (SUS316L).
The magnetism-pumped circular water current can heat the system evenly, and the bath liquid can achieve an equal temperature.
A MPU is used in dissolution tester to automatically control the temperature, rotational speed and time, and the parameters can be preset at any time.
It can be operated conveniently by keystroke. The measurement result is displayed by LED. The preset and real data can be displayed alternately in time.
Automation: auto-test, auto-diagnose, auto-alarm.
It can test by self and alarm automatically. The dissolution parameters of'Chinese codex 2000'are stored in and can be taken out at anytime.
Time can be preset at many points. The beeper sounds automatically to any point, and then the tester automatically runs to the next one.
Speed range (20~200)rpm
Speed accuracy ±2rpm
Temperature range ambient to +45.0°C
Preset timing nine points (1~999)min
Time accuracy ±0.5mm
Paddle radial run-out ±0.5mm
Basket radial run-out ±1.0mm
Power 220V/50Hz/1200W or 110V/60Hz/1200W
Our Packaging Details:
Q: How long is the delivery time?
A: 7-15 working days after we received your payment or order confirmed.
Q: What is the warranty for the equipment?
A: One year. We can provide you lifelong technical support.
Q: What is the payment terms?
A: T/T, L/C, Western Union, etc.
Q: What after-sales service can be provided by Nade ?
A: The after-sales service of Nade is divided into two parts: technical support and maintenance.
Q: What about the technical support and maintenence?
A: Sometimes users will get some problems when they operate the instruments which are unfamiliar or first time use, users can send us emails or call us about detailed information. We will provide users solution or operation video.For so many years’ experience, our instruments works well during warranty period. If there any parts are broken during warranty period, we would like to provide spare parts for free.
Please send your request details to us by clicking "send" at the bottom!